multifunction face raised facial hifu wrinkles 3D Hifu Beauty Machine 3DV
1. DS-1.5 used to transmit high-energy focused ultrasound to the epidermis tissue, the ultrasound to penetrate the skin to a depth of 1.5mm, is responsible for activating the skin epidermis in thinner tissue.
2. Transducer DS-3.0mm, 4MHz frequency ,used to release ultrasound energy into the dermis layer of 3.0mm,is responsible for activating the skin’s dermal layer of collagen, effectively enhance the effect of the consolidation of the outline, but also improve large pores and reduce the appearance of wrinkles.
3.Transducer DS-4.5mm, 4MHZ frequency used to transmit high- energy focused ultrasound to the skin, subcutaneous tissue, the ultrasound to penetrate the skin to a depth of 4.5mm, direct subcutaneous SMAS layer, forming “heat coagulation” regional, targeted to thicker skin, such as cheeks, etc.
HIFU machine is an advanced new high-intensity focused ultrasound technology designed instrument, change the traditional face lift wrinkle cosmetic suigery, non-surginal wrinkle technology, the Hifu machine will release high concentrated focus sonic energy can penetrate in the deep SMAS fascia skin tissue and coagulation of high heat in the proper position, the deep dermis to stimulate the skin to produce more collagen and thus to tighten so the skin becomes former. hifu can directly delivers heat energy to skin and subcutaneous tissue that can stimulate and renew the skin's collagen and thus consequently improving the texture and reducing sagging of the skin. It literally achieves the results of a facelift or a body lifts without any invasive surgery or injections, moreover, an added bonus of this procedure is that there is no downtime.
multifunction face raised facial hifu wrinkles 3D Hifu Beauty Machine 3DV.Each cartridge has 15,000 shots.Painless, Effectively,High-quality.
◆Clean the Skin.
◆Take a photo before treatment , and paint on the preoperative lifting location with make - up pencil .
◆According to the skin thickness to select appropriated transducer (Tranmitter) .
◆Set suitable input intensity ( such as1.0-1.5J ) , focal length and spacing ( such as 1.5mm)
◆Applying ultrasonic plentiful gel as couplant between transducer and skin.
◆face lifting operate on the note of power line, forming a line thermal condensed node in the specified depth under skin when the transmitting button is pressed per time,operating continued on the powder line until the specified location transmission is completed.
◆Don’t operate on marginal mandibular nerve and supra or bital nerve.
◆No recovery period after treatment,daub some moisture product and make up immediately.
◆Up to 11 lInes can be ht In one shot, and the maxlmum energy area wldth Is 10mm. AccordIng to the skin area of Page4
Page5 AdJust the correspondng parameters of the sIze, greaty shorten the operatlon time, make the energy polnts on the skn
more unlform, and the effect Is better.
◆The use of the most advanced high-tech technology, equlpped with two treatment heads accordIng to the faclal skn condl-
tlon, accurately acting on different skIn depths, energy Is sklpped over the epldermls durng treatment, 100% wthout any
damage. At the same tme, the depth of the skin of the treatment head Is consistent with the set value to ensure that the custom-
er Is palnless and comfortable.
◆The thermal efect on dermal collagen and colagen flbers also has thermal stmulaton to the fat layer and fasda layer (SMAS),
and the therapeutlc effect Is far better than that of thermal maggle.
WHAT OUR CLIENTS SAY
What to do after treatment？
◆Avoid exercise that will create friction and make sure to gently pat off any sweat
or water to dry the skin under your dressing.
◆For the first 3 days after treatment, keep the treated area bandaged with sterile
gauze and protected with a thin layer of antibiotic or healing ointment.
◆After 3 days, keep the treated area clean and dry during the healing processIt
is recommended to wait at least six weeks after gettinga new tattoo to begin
the laser tattoo removal process.
◆Don’t eat Spicy food within 1week
◆3D HIFU machine with 11 lines per shot, one step 10MM treatment width2.
◆Each cartridge has 15,000 shots
◆One shot is total 11 lines and total 25mm length ,this hifu beauty machine deliever the hifu energy to skin layer from 1.5mm to 16mm more precise ,stable, and faster.
Equipment daily maintenance
Check Voltage normal before Operate.
Keep working Room constant temperature and dry.
Wipe Lens ,Wipe crystal ,removal gel ,keep machine clean.
Next page >>
Approvred vaginal tightening facial lifting wrinkle removal 3d HIFU beauty equipment 3D HIFU 300BView More >
4Hz 8 Cartridges 3D HIFU Machine For Facial And Body Lifting Equipment 3D HIFU 300View More >
Home use skin rejuvenation wrinkle removal focused HIFU beauty machine HF119View More >
The Global Beauty Group is registered under the China Cooperation Act,2008 Business Number 91110115674295685F.